The client has a problem with its elevating leg rest system where his legs slip between the calf supports and footrests. Also the head support is inadequate. The wheelchair was modified by fitting an upholstered panel and one piece footrest, bridging the gap and locking the legrests and footrests together. The chest/head support frame was replaced by a horizontal square MS tube with height and lateral adjustment, incorporating a locking facility for the adjusting straps.
Bristol panel case ref 11\15-06